Creating a free account will enable you to subscribe to our daily and weekly email newsletters, as well as customize your reading experience to show only the categories most relevant to you.
Signing up only take a few minutes, so why not give it a try and see what you've been missing out on.
Certain HeartStart automated external defibrillator (AED) devices made by Philips Medical Systems, a division of Philips Healthcare, may be unable to deliver needed defibrillator shock in a cardiac emergency situation, the U.S. Food and Drug Administration said on December 3rd in a new safety communication for users of these previously recalled devices.
The safety communication includes recommendations to better inspect and monitor the readiness of these devices, as well as steps to follow if someone must use a recalled device in an emergency situation.
"The FDA advises keeping all recalled HeartStart AEDs in service until you obtain a replacement from Philips Healthcare or another AED manufacturer, even if the device indicates it has detected an error during a self-test," said Steve Silverman, director of the Office of Compliance in the FDA's Center for Devices and Radiological Health. "Despite current manufacturing and performance problems, the FDA considers the benefits of attempting to use an AED in a cardiac arrest emergency greater than the risk of not attempting to use the defibrillator."
These devices were manufactured and distributed between 2005 and 2012 under the names HeartStart FRx, HeartStart HS1 Home, and HeartStart HS1 OnSite. Users, who may include consumers and first responders, should contact Philips Healthcare immediately for a replacement AED unit.
An AED is a device that automatically analyzes the heart rhythm in victims of sudden cardiac arrest and delivers an electrical shock to restore its normal rhythm. AEDs help save lives of cardiac arrest victims when they are working properly and used correctly. Each year, nearly 300,000 Americans collapse from sudden cardiac arrest. When normal heart rhythms are not restored quickly, sudden cardiac arrest can cause death.
In September 2012, Philips Healthcare initiated the recall of HeartStart FRx, HeartStart HS1 Home, and HeartStart HS1 OnSite AEDs due to the failure of an internal electrical component. The recall affected approximately 700,000 devices. In a Medical Device Safety Notice dated November 19, 2013, Philips provided consumers with updated information about the failure of an internal electrical component that could cause the AEDs to fail to deliver a shock. The notification also directed users to a Maintenance Advisory.
The FDA will continue to closely monitor all AED manufacturers' quality system practices and manufacturing changes that have persistently contributed to recall and adverse events associated with AEDs.
In March 2013, the FDA issued a proposed order that if finalized would require manufacturers of AEDs and accessories to submit premarket approval applications that focus specifically on the critical requirements necessary to assure AEDs are safe and effective. The main objective of this proposed regulatory approach is to improve the reliability of AEDs so that they can continue to save lives.
Article adapted by Medical News Today from original press release. Click 'references' tab above for source.
Visit our Medical Devices / Diagnostics category page for the latest news on this subject.
Please use one of the following formats to cite this article in your essay, paper or report:
FDA. "FDA issues safety communication on HeartStart automated external defibrillators from Philips Healthcare." Medical News Today. MediLexicon, Intl., 5 Dec. 2013. Web.
10 Mar. 2014. <http://www.medicalnewstoday.com/releases/269694>
FDA. (2013, December 5). "FDA issues safety communication on HeartStart automated external defibrillators from Philips Healthcare." Medical News Today. Retrieved from
Please note: If no author information is provided, the source is cited instead.
If you write about specific medications, operations, or procedures please do not name healthcare professionals by name.
For any corrections of factual information, or to contact our editorial team, please use our feedback form. Please send any medical news or health news press releases to:
Note: Any medical information published on this website is not intended as a substitute for informed medical advice and you should not take any action before consulting with a health care professional. For more information, please read our terms and conditions.
This page was printed from: http://www.medicalnewstoday.com/releases/269694.php
Visit www.medicalnewstoday.com for medical news and health news headlines posted throughout the day, every day.
© 2004-2014 All rights reserved. MNT (logo) is the registered trade mark of MediLexicon International Limited.